Model 1154 Precision Conductivity Meter. In standard form measures conductivity in 3 ranges via 3 buttons and temperature via 1 button.
Button 1 - 0-20 pS/m, resolution 0.01 pS/mButton 2 - 0-200 pS/m, resolution 0.1 pS/mButton 3 - 0-2,000 pS/m, resolution 1 pS/mButton 4 - Temperature
For use with ASTM D4308, D2624
Many applications require highly accurate “rest” electrical conductivity measurements of fluids ranging in values from less than one pS/m to several thousand. “Rest” conductivity is the electrical conductivity at the initial instant of current measurement after a dc voltage is impressed between electrodes (reciprocal of the resistance of uncharged fluid in the absence of ionic depletion or polarization).
The Precision Conductivity Meter was originally designed to precisely measure the conductivity of hydrocarbon fuels only. Limiting factors are the viscosity of the fluid and the conductivity level to be measured. However, electrical conductivity of other fluids can be measured and/or ranges of the instrument can be modified by Emcee to accommodate higher conductivity values than the standards offered.
Theory of Operation
A specific volume (100-mL) of test sample is introduced between the two concentric electrodes in the Cell. Using a triaxial cable, the Cell is connected to a sensitive dc ammeter in the Console and a dc voltage is applied between the two electrodes. Due to the physical geometry of the Cell, the Cell constant is stable and calculable. Thus, the resultant peak current is readily transformed and digitally displayed as picosiemens per meter (pS/m), also commonly known as conductivity units (CU). As defined in ASTM Standard Test Method D 4308, a CU is equal to a picosiemen per meter (1 CU = 1 pS/m). The enclosure technique employed in the Cell precludes the need for external screening from stray electrical fields.
Agility always receive a 1st class service from Med-Lab, we would not deal with any other Dangerous Goods packer.
Sally O - Castle Donnington, Derbyshire